ok I have an HR2510 that had some issues so I recapped it and removed, cleaned and checked all components that had the corrosive glue on them. I am still seeing the same issues, no display for the frequency but I can see the bar meter work to the left and the channel up & down buttons along with the frequency dial are inoperable. I am able to change channels via the microphone buttons. So now that I got past it not being a bad cap or corroded Component we’re should I start looking? I was going to lift D152 to check it but it cracked, now I don’t know if it was cracked or me desoldering it cracked it? So that’s on ordered for replacement. I also pulled D112 &113 and checked on Fluke, 112 shows .337 and OL and 113 shows .609 and OL. I don’t show power on either side of it but I believe it gets power from D152 so that will need to wait. Am I on the right path or should I be looking elsewhere? Any tips or suggestions?